The mountaineer's method for peak detection in photoplethysmographic signals
Keywords:photoplethysmography, peak detection, low-amplitude signals, noise contamination, adaptive threshold
Several efforts have been made to develop algorithms for accurate peak detection in photoplethysmographic (PPG) signals. Most of those algorithms have been specifically conceived to perform under high motion artifact and baseline drift conditions. However, little has been done regarding peak detection in low-amplitude PPG signals. In an attempt to address this issue, a simple and real-time peak detection algorithm for PPG signals was proposed. In comparison with two other well-established peak detection algorithms, the proposed method was able to achieve over than 98% sensitivity and less than 3% failed detection rate, even when the amplitude of the PPG signal dropped to 0.2 V. Still, further work is needed to improve its robustness to motion artifacts.
J. Allen, “Photoplethysmography and its application in clinical physiological measurement,” Physiological Measurement, vol. 28, no. 3, pp. R1–39, Feb. 2007.
R. Colombo and et al., “Pulse photoplethysmographic analysis estimates the sympathetic activity directed to heart and vessels,” Anesthesiology, vol. 123, no. 2, pp. 336–345, Aug. 2015.
M. Elgendi, “On the analysis of fingertip photoplethysmogram signals,” Current Cardiology Reviews, vol. 8, no. 1, pp. 14––25, Feb. 2012.
K. Hamunen and et al., “Effect of pain on autonomic nervous system indices derived from photoplethysmography in healthy volunteers,” British Journal of Anaesthesia, vol. 108, no. 5, pp. 838–844, May 2012.
W. J. Jiang, P. Wittek, L. Zhao, and S. C. Gao, “Adaptive thresholding with inverted triangular area for real-time detection of the heart rate from photoplethysmogram traces on a smartphone,” in 36th Annual International Conference of the IEEE Engineering in Medicine and Biology Society (EMBC), Chicago, IL, 2014, pp. 3212–3215.
W. Karlen, J. M. Ansermino, and G. Dumont, “Adaptive pulse segmentation and artifact detection in photoplethysmography for mobile applications,” in 34th Annual International Conference of the IEEE Engineering in Medicine and Biology Society (EMBC), San Diego, CA, 2012, pp. 3131–3134.
Z. Marcinkevics, S. Kusnere, J. I. Aivars, U. Rubins, and A. H. Zehtabi, “The shape and dimensions of photoplethysmographic pulse waves: a measurement repeatability study,” Acta Universitatics Latviensis Biology, vol. 753, pp. 99–106, 2009.
R. C. Peng, X. L. Zhou, W. H. Lin, and Y. T. Zhang, “Extraction of heart rate variability from smartphone photoplethysmograms,” Computational and Mathematical Methods in Medicine, vol. 1, pp. 1–11, 2015.
K. H. Shelley, “Photoplethysmography: beyond the calculation of arterial oxygen saturation and heart rate,” Anesthesia & Analgesia, vol. 105, no. 6, pp. S31––S36, Dec. 2007.
H. S. Shin, C. Lee, and M. Lee, “Adaptive threshold method for the peak detection of photoplethysmographic waveform,” Computers in Biology and Medicine, vol. 39, no. 12, pp. 1145–1152, Dec. 2009.
T. V. Tran and W. Y. Chung, “A robust algorithm for real-time peak detection of photoplethysmograms using a personal computer mouse,” IEEE Sensors Journal, vol. 15, no. 8, pp. 4651–4659, Aug. 2015.
T. V. Tran and W. Y. Chung, “A robust peak detection algorithm for photoplethysmographic waveforms in mobile devices,” Journal of Medical Imaging and Health Informatics, vol. 7, no. 7, pp. 1617–1623, Nov. 2017.
D. Žikić, “An improved reflective photoplethysmograph probe design for detection of an arterial blood flow,” Journal of Medical Engineering & Technology, vol. 32, no. 1, pp. 23–29, Jan. 2008.
How to Cite
Copyright (c) 2019 Revista Facultad de Ingeniería Universidad de Antioquia
This work is licensed under a Creative Commons Attribution-NonCommercial-ShareAlike 4.0 International License.
Revista Facultad de Ingeniería, Universidad de Antioquia is licensed under the Creative Commons Attribution BY-NC-SA 4.0 license. https://creativecommons.org/licenses/by-nc-sa/4.0/deed.en
You are free to:
Share — copy and redistribute the material in any medium or format
Adapt — remix, transform, and build upon the material
Under the following terms:
Attribution — You must give appropriate credit, provide a link to the license, and indicate if changes were made. You may do so in any reasonable manner, but not in any way that suggests the licensor endorses you or your use.
NonCommercial — You may not use the material for commercial purposes.
ShareAlike — If you remix, transform, or build upon the material, you must distribute your contributions under the same license as the original.
The material published in the journal can be distributed, copied and exhibited by third parties if the respective credits are given to the journal. No commercial benefit can be obtained and derivative works must be under the same license terms as the original work.